<p>&nbsp;<strong style="font-family: OpenSans-Regular; font-size: 14px; text-align: justify; box-sizing: border-box;">&nbsp;AEROSENSE TRANSMITTER</strong></p>
<p style="box-sizing: border-box; margin: 1.5em 0px 0px; padding: 0px; font-size: 14px; line-height: 2em; text-align: justify; font-family: OpenSans-Regular;"><span style="box-sizing: border-box; color: rgb(104, 104, 104); font-family: roboto, Helvetica, Tahoma, Geneva, Arial, sans-serif; font-size: 16px;">The Model ECMT200 Carbon Monoxide Transmitter provides a field selectable current or voltage output that is proportional to the gas concentration in underground parking garages, vehicle maintenance facilities, or mechanical rooms. A field replaceable, electrochemical sensor provides accurate readings for up to 4 years with proper calibration</span></p>
<p style="box-sizing: border-box; margin: 1.5em 0px 0px; padding: 0px; font-size: 14px; line-height: 2em; text-align: justify; font-family: OpenSans-Regular;">&nbsp;</p>
<ul style="box-sizing: inherit; margin: 1.5rem 0px 1.5rem 1.5rem; padding: 0px; text-align: center; color: rgb(104, 104, 104); font-family: roboto, Helvetica, Tahoma, Geneva, Arial, sans-serif; font-size: 16px;">
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Sensor:</strong>&nbsp;Field replaceable electrochemical, 4 year typical lifespan. Range: 0 to 200 ppm.</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Output Drift:</strong>&nbsp;&lt;5% per year in air.</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Coverage Area:</strong>&nbsp;5000 to 7000 sq. ft. typical.</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Accuracy:</strong>&nbsp;&plusmn;2% of reading at the time of calibration.</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Temperature Limits:</strong>&nbsp;-4 to 122&deg;F (-20 to 50&deg;C).</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Storage Temperature:</strong>&nbsp;For best sensor life, 32 to 68&deg;F (0 to 20&deg;C). Humidity Limits: 15 to 90% RH constant; 0 to 99% RH intermittent. Response Time: &lt;45 seconds to 90% of final value.</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Calibration:</strong>&nbsp;15 turn span and zero adjustment potentiometers. Housing: UV resistant polycarbonate.</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Output:</strong>&nbsp;Jumper selectable 4 to 20 mA (loop powered) or 2 to 10 V (load must be &gt;50 KQ).</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Power Requirements:</strong>
    <ul style="box-sizing: inherit; margin: 0px 0px 0px 1.5rem; padding: 0px;">
        <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;">Current Output: 18 to 28 VDC;</li>
        <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;">Voltage Output: 18 to 28 VDC/VAC, reverse polarity protected.</li>
    </ul>
    </li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Electrical Connection:</strong>&nbsp;Removable terminal block, includes two PG11and one PG 16 knockouts for conduit fitting.</li>
    <li style="box-sizing: inherit; text-align: justify; line-height: 30px; margin-top: 15px; margin-left: 16px;"><strong style="box-sizing: inherit;">Weight:</strong>&nbsp;0.28 lb (0.11 kg)</li>
</ul>