<div class="pll" style="font-family: verdana; text-align: justify; color:white;">Having vast experience in the relevant industry, we are engaged in providing a broad assortment of&nbsp;<b>Galaxy Magnehelic Gauge</b>&nbsp;that is assemble in order to meet specific demands of customers. Moreover, a series of quality tests is performed on offered gauge to ensure a defect free delivery at clients end. Clients can avail the offered range from us at market leading prices within stipulated time frame.</div>
<div class="pll" style="font-family: verdana; text-align: justify; color:white;"><br />
Features:<br />
<br />
Accuracy<br />
High functionality<br />
Long service life<br />
<br />
Specifications:<br />
<br />
We provide&nbsp;<b>Galaxy magnehelic gauge</b>&nbsp;of series which is low in cost, diaphragm operated and differential pressure gauge<br />
Owing to these features these are used in applications for measuring positive, negative, or differential pressure with an accuracy of 2%<br />
Besides, these are also applied for monitoring filter status, duct static pressure, room pressure, fan or blower pressure, paint booths, dust collectors, cabinet purging and many others<br />
We offer Galaxy magnehelic differential pressure gauges which run on air and non-combustible compatible gases and are offered with the option of natural gas<br />
In addition to this, these housing are fabricated by using die cast aluminum case and bezel, with acrylic cover<br />
These have exterior finish with grey color to withstand 168 hour salt spray corrosion test<br />
<br />
Technical Specifications:<br />
<br />
<br />
Accuracy: &plusmn;2% of full scale (&plusmn;3% on - 0, -100 Pa, -125 Pa, 10MM and &plusmn;4% on - 00, -60 Pa, -6MM ranges), throughout range at 70&deg;F (21.1&deg;C)<br />
Pressure Limits: -20 Hg. to 15 psig.&dagger; (-0.677 bar to 1.034 bar); MP option<br />
HP option: 35 psig (2.41 bar), 80 psig (5.52 bar)<br />
Overpressure: Relief plug opens at approximately 25 psig (1.72 bar), standard gages only<br />
Temperature Limits: 20 to 140&deg;F.* (-6.67 to 60&deg;C)<br />
Size: 4 (101.6 mm) Diameter dial face<br />
Mounting Orientation: Diaphragm in vertical position, Consult factory for<br />
Process Connections: 1/8 female NPT duplicate high and low pressure taps one pair side and one pair back<br />
Weight: 1 lb 2 oz (510 g), MP &amp; HP 2 lb 2 oz (963 g)<br />
Other position orientations<br />
<br />
Applications:<br />
<br />
Filter Status<br />
Duct Static Pressure<br />
Clean Room Pressure<br />
Building Pressure<br />
Fan and Blower Pressure<br />
Paint Booths<br />
Dust Collectors<br />
Glove Box Pressure<br />
Bubbler Systems<br />
Cabinet Purging</div>
<div class="pll" style="font-family: verdana; text-align: justify;"><br />
<p class="c26 b" style="margin: 0px; padding: 0px; line-height: 20px; font-weight: bold;"><br />
Additional Information:</p>
<ul style="margin: 0px 0px 0px 0px; padding: 0px; line-height: 20px !important;">
    <li style="margin: 0px; padding: 0px; line-height: 20px !important;"><b>Minimum Order Quantity:&nbsp;</b>9650264867 Cell Us</li>
    <li style="margin: 0px; padding: 0px; line-height: 20px !important;"><b>Pay Mode Terms:&nbsp;</b>T/T (Bank Transfer)</li>
</ul>
</div>